Covid-19 Antibody Nasal Swab Saliva Test Kit In Vitro Diagnostic Reagent
Contact Person : Zhang
Phone Number : 86-13392649067
WhatsApp : +8613392649067
|Minimum Order Quantity :||1000 pcs||Packaging Details :||10 PCS / BOX,|
|Delivery Time :||5-8 work days,according to order quantity||Payment Terms :||L/C, T/T, Western Union|
|Supply Ability :||100000 pcs / day|
|Place of Origin:||Shenzhen,Guangdong||Brand Name:||Lycome|
|Self Life:||2 Years||Test Item:||Covid-19|
|Reading Time:||10-15 Min||Accuracy:||99%|
|Specimen:||Saliva||Package:||1 Pcs / Pouch|
|Instrument Classification:||Class III||OEM Service:||Acceptable|
|Application:||Home & Medical Examination|
Accurate Rapid Antigen Test Kit For Respiratory Disease Coronavirus Saliva Specimen Test
The COVID-19 Ag Rapid Test is an in vitro immunochromatographic assay for the qualitativ detection of Influenza COVID-19 antigen in a test device from nasal swab, throat swab or nasal wash/aspirate specimens. It is intended to aid in the rapid differential diagnosis of influenza SARS-CoV-2 viral infections. This test provides only a preliminary test result. Therefore, any reactive specimen with the COVID-19 Ag Rapid Test must be confirmed with alternative testing method(s) and clinical findings. The test is intended for professional and laboratory use.
Test Device (25): Each foil pouch contains one single-use test device with one membrane strips. The strip contains a test line of monoclonal antibody
Extraction Tube(25): One tube containing 0.5ml extraction solution.
Sterile Nasal Swab(25)
1. Store at room temperature (2-30℃ or 35.6-86°F) in a dry place. Avoid direct sunlight;
2. 12 months of shelf life (date of manufacture to expiration date).
Step 1 Bring the test to room temperature (18-26℃) before use.
Step 2 Open the foil pouch, take out the test and lay it on an even surface.
Step 3 Add 5 drops(about 80 µL )of extraction mixture into the each sample well (S).
Step 4 Read and record the results at 15 minutes after addition of samples.
NOTE: Result will be invalid after 20 minutes. After observing and recording the result, dispose of bio-hazardous materials by following the practices of your institution. Discard all the materials in a safe and acceptable manner in compliance with all the federal, state, and local requirements.
【EPLANATION OF TEST RESULTS】
1. Negative result: if only quality control line C appears and test line T show no color, it means no Influenza COVID-19 antigen is detected and the test result is negative (as shown in the following figure).
2.Positiveresult:If the third strip both quality control line C and T appear, it means the COVID-19 antigen is detected and the test result is the COVID-19 antigen positive (as shown in the following figure)
3. Invalid result: if quality control line C can not be observed, regardless of test line, the result would be invalid.(as shown in the following figure) The sample should be tested again.
1.The intensity of color in the test line (T) may vary depending on the concentration of analyses present in the specimen. Therefore, any shade of color in the test line (T) should be considered positive. Please note that this is a qualitative test only, and cannot determine the concentration of analyses in the specimen.
2.Insufficient specimen volume, incorrect operating procedure or expired tests are the most likely reasons for control band failure.
1. Negative reference coincidence rate
Use 10 samples of COVID-19 Ag negative reference to test, the test results should not be positive, the negative coincidence rate should be 10/10；
2. Positive reference coincidence rate
Use 10 samples of COVID-19 Ag positive reference to test, the test result should all be positive, the positive coincidence rate should be 8/10;
Use 3 samples of COVID-19 Ag sensitivity reference to test, L1 should be detected, L2 should be detected or not,L3 should not be detected.
Use 1 sample of internal control repeatability reference( J1) to complete the parallel test for 10 tests, the results should be coincident.
5. Cross-reaction and interference reaction
Use positive samples of Rhinovirus, RS virus, Influenza A virus, Influenza B virus, Chlamydia, Mycoplasma and Bacterial Infection to do the tests, the test results should be negative.
6. The Clinical study
The COVID-19 Ag Rapid Test has been evaluated with 155 clinical samples. Both 91 negatives and 64 positives were confirmed by PCR
The COVID-19 Ag Test v.s. PCR
|Viral culture results|
|The COVID-19 Ag Test||Positive||52||7||59|
 Sensitivity = A/(A＋ C) ×100％＝ 81.3%
 Specificity = D/(B＋ D) ×100％＝ 92.3%
 Crude coincidence rate= (A+D)/(A+B+C+D)×100%＝ 87.7%
 Consistency factor kappa = 0.744
【EXPLANATION OF SYMBOLS ON LABEL AND PACKAGE】
Lycome International Industry Limited was established in 2020 and is located in Zhongshan City, Guangdong Province, with a registered capital of 10 million yuan. We have our own large-scale production base which covered more than 6 production line,10000 square meters workshop.
Lycome International Industry Limited is a high-tech private enterprise dedicated to the research, development, production, sales and service of medical consumable products. We insist on the aim of "Creation, development and effective communication"Aswell as built the spirit of teamwork with "Human health, concerning on service". We've formed a complete one-stop service systems include R & D, production, sales, logistics, after-sales service.
Enter Your Message